|Published (Last):||24 August 2005|
|PDF File Size:||12.63 Mb|
|ePub File Size:||3.20 Mb|
|Price:||Free* [*Free Regsitration Required]|
Quick Links. Download this manual. Table of Contents. This instruction manual contains warnings and safety rules which must be observed by the user to ensure safe operation of the instrument and retain it in safe condition. Page 4 Following symbols are used on the instrument and in the instruction manual. Attention should be paid to each symbol to ensure your safety. Refer to the instructions in the manual. Return the instrument to Kyoritsu or your distributor for repair or re-calibration. Higher-numbered categories correspond to electrical environments with greater momentary energy, so a measuring instrument designed for CAT III environments can endure greater momentary energy than one designed for CAT II.
The effective value is the square root of the average of square of alternating current or voltage values. Many clamp meters using a conventional rectifying circuit have "RMS" Press to open the Transformer Jaws. The instrument defaults to the normal mode NOR. Then, press this switch to cycle through measuring modes. In any mode, pressing this switch for more than one second returns the instrument to the normal mode. This may cause shock hazard or damage to the instrument or equipment under test.
Otherwise, accurate measurement cannot be made. The maximum measurable conductor size is approx. The most accurate reading will be obtained by keeping the conductor at the center of the transformer jaws.
The buzzer beeps, if the resistance under test is This may cause electrical shock hazard and damage to the instrument or the circuit under test. Make sure to set the function selector switch to the OFF position after use. This is a function to prevent the instrument from being left powered on in order to conserve battery life.
Make connection to the recording device. Set the range depending on the state of Data Hold Switch. Print page 1 Print document 33 pages. Cancel Delete. Sign In OR. Don't have an account? Sign up! Restore password. Upload from disk. Upload from URL.
Kyoritsu AC/DC Digital Clamp Meters KEW 2009R
KYORITSU KEW SNAP 2009R Instruction Manual